ISO 10555-5 EBOOK
ISO. Second edition. Reference number Permission can be requested from either ISO at the address below or ISO’s member body in the. 31 Jul Purchase your copy of BS EN ISO as a PDF download or hard copy directly from the official BSI Shop. All BSI British Standards. EN ISO Intravascular catheters – Sterile and single-use catheters – Part 5 : Over-needle peripheral catheters – ISO specifies requirements.
|Published (Last):||19 March 2016|
|PDF File Size:||1.90 Mb|
|ePub File Size:||9.3 Mb|
|Price:||Free* [*Free Regsitration Required]|
Other parts of this 10555- are as follows: The details of conditions under which the licence for the use of the Standard Iso 10555-5 may be granted iso 10555-5 manufacturers or producers may be obtained from the Bureau of Indian Standards. E-5 Test report The test report shall contain at least the following information: Certain conventions isl, however, not identical to those used in Indian Standards.
Page 3, subclause 4. Available from Munsell Color, N.
Search the history of over billion web pages on the Internet. Until that time, a manufacturer may label his product iso 10555-5 opaque” provided iso 10555-5 can support this claim by demonstrat- ing that he has an appropriate method for showing radio- opacity.
The corresponding Indian Standards which are to be substituted in their places are listed below along with their degree of iso 10555-5 for the editions indicated: Designation and catalogue of CCR colours. The fluid pathway in the needle shall be free of unintended obstructions that would prevent flashback.
Part 1 General requirements do The technical committee has reviewed the provisions of the following International Standard referred in this adopted standard and has decided that it is acceptable for use in conjunction with this standard: Attention is particularly drawn to the following: This Indian Standard has been developed from Doc No.: Bureau of Iso 10555-5 Standards Iso 10555-5 is a statutory institution established under the Bureau of Iso 10555-5 Standards Act, to promote harmonious development of the activities of standardization, marking and quality certification of goods and attending to connected matters in the country.
International Standard ISO Blood flow into the needle hub.
An approved test method for iso 10555-5 a value of radio-detectability will be established. Iso 10555-5 part of the these publications may be reproduced in any form without the jso permission in writing of BIS. Measure the time taken for fluid to form the first falling drop at the back of the vent fitting.
Annex D shows examples of typical needle point geom- etries. Peripheral intravascular catheter designed iso 10555-5 the introduction of withdrawal of liquids or devices into or from the peripheral vascular system. NOTE To improve the iso 10555-5 of the solution, a colorant such as red or blue food dye may be incorporated.
iso 10555-5 Standards are also reviewed periodically; a standard alongwith amendments is reaffirmed when such review indicates that no changes are needed; if iso 10555-5 review indicates that changes iso 10555-5 needed, it is taken up for revision.
Suggested colour references for opaque naaterials are given in annex C. Assembly comprising the cath- eter tube, catheter hub and any integral fittings. Pages 1 and 2 4. The number of significant places retained in the rounded off value should be the same as that of the specified value in this standard.
Intravascular catheters ISO , -3, -4, -5
The catheter iso 10555-5 may have a single lumen or multiple lumens. Limited collection of colours. Copyright BIS has the copyright of all its publications. Details of which is given in National Annex A. NOTE 3 The point should be designed to be noncoring. Review of Indian Standards Amendments are issued to standards iso 10555-5 the need arises on iiso basis of comments.
BS EN ISO 10555-5:2013
Fitting attached to the needle tube, providing communication with its bore. In this adopted standard, reference appears to iso 10555-5 International Standards for which Indian Standards also exist.
When the needle is fully inserted into the catheter unit, the catheter tube shall neither extend iso 10555-5 the isp of the needle bevel nor be iso 10555-5 than 1 nnnn fronn it see dinnension a in figure 1.
All standards are subject to revision, and parties to agreements based on this part of ISO are encouraged to investigate the possibility of applying the most recent editions of the standards indicated below.